The ref cavity ion pump was running at 7.7kV instead of 5kV
This Digitel SPC-1 20 l/s ion pump should be running at 5kV
I noticed that the ion pump was turned off.
It was turned ON. It showed 0.00 microA at 5kV The current display is not sensitive enough. There must be some small outgassing or leak. It adds up if we stop pumping.
We want to keep the reference cavity in pristine condition. It required the ion pump running all times.