Not logged in
Sun Jul 13 17:19:29 2008
MOPA SLOWM Calibration
We first unlocked the FSS and ramped the SLOW actuator. With the PMC locked we observed the PMC PZT voltage
as a function of SLOWM (SLOW loop actuator voltage). We believed this to be ~1-5 GHz / V. Since this is
not so precise we then ran a slow (2 min. period) triangle wave into the slow actuator and looked at the
ref cav transmission peaks to calibrate it.
Plot is attached>
We assume that the
reference cavity length = 203.2 mm
then the FSR = 737.7 MHz. So looking at the plot
and using our eye to measure the
SLOWM calibration is 1054 +/- 30 MHz/V
. This is probably dominated by
our eye method.
Note: we tried to get the length from T010159-00-R (Michele, Weinstein, Dugolini). In that doc,
the length used is 203.3 mm whereas its 203.2 mm in the PSL FDD (?). The calculation of the FSR is also
incorrect (looks like they used c = 299460900 instead of 299792458 m/s). We took the length from the PSL FDD
(T990025-00-D) but not the FSR, since they also did not find the right value of 'c'. I guess that the speed
of light just ain't what it used to be.