I've finished using the network analyzer to characterize find a dither frequency for driving the PZT to use in my PDH locking. I found a region in which the amplitude response of the PZT is low: The dip is centered at 2.418 MHz. Changing the NPRO laser temperature by 100mK has no significant effect on the transfer function in that region. I will post plots tomorrow.
I'm finished with the network analyzer. It is unplugged, and the cart is still near the PSL table. (I'll roll it back tomorrow when it won't disturb interferometer locking).
I closed the shutter on the NPRO at the end of the night.
Tomorrow I plan to put together the fast locking setup. I'll drive the PZT at 2.418 MHz. More details to come tomorrow.