The power of the beam which is being coupled into the optical fibre is measured to be between 159 mW to 164 mW (The power meter was showing fluctuating readings).
The power out of the beam coming out of the fibre far-end at the PSL table is measured to be 72 mW. Here, I have attached a picture of the beam paths of the ETMY table with the beams labelled with their respective powers.
For the phase locking or beat note measuring we only need ~1 mW. Its a bad idea to send so much power into the fiber because of SBS and safety. The power should be lowered until the output at the PSL is < 2 mW. In terms of SNR, there's no advantage to use such high powers.