The good news is that we seem to be running in a linear region of the PSL laser with a degree or so of range before the PSL Innolight laser starts to run multi-mode. On the attached graph we are currently running the PSL at 32.26degrees (measured) which puts us in the lower left corner of the plot. The blue data is the Lightwave set temperature (taken from the display on the laser controller) and the red data is the Lightwave laser crystal measured temperature (taken from the 10V/degC calibrated diagnostic output on the back of the laser controller - between pins 2 and 4).
The other good news is that we can see the transition between the PSL laser running in one mode and running in the next mode along. The transition region has no data points because the PMC has trouble locking on the multi-mode laser output - you can tell when this is happening because, as we approach the transition the PMC transmitted power starts to drop off and comes back up again once we're into the next mode region (top left portion of the plot).
The fitted lines for the region we're operating in are:
Y_arm_Temp_meas = 0.95152*T_PSL + 3.8672
Y_arm_Temp_set = 0.87326*T_PSL + 6.9825