[Rana and Kevin]
I measured the optical transfer function of POY and fit the data using LISO. The fit can be found at http://lhocds.ligo-wa.caltech.edu:8000/40m/Electronics/POY. POY was missing the RF cage and back cover so I took those parts from AS55 in order to make these measurements.
POY does not have the unwanted oscillations at 225 MHz that POX has. Attachment 1 shows the transfer functions of POX and POY.
To measure the transfer functions, I used a 50/50 beam splitter to send half the light from an AM laser to POY and half the light to a New Focus 1611 reference photodiode. The transfer function for POY was measured as the transfer function of the signal from POY divided by the signal from the 1611. When I was measuring the transfer function for POX, I failed to ensure that the photodiodes were operating linearly. Before making the measurements for POY, I varied the RF power modulating the AM laser and recorded the magnitude of the transfer function at the 11 MHz peak. Attachment 2 shows these values. The measurements for POY were made in the linear region at an RF power of -10 dBm. The measurements for POX were made at 0 dBm and were most likely not in the linear region for POX.