Not logged in
Mon Dec 20 10:47:29 2010, Koji, Update, IOO, Plan for closing the vacuum chambers
Mon Dec 20 16:57:58 2010, steve, Update, IOO, checking out & closing the vacuum chambers
Mon Dec 20 23:32:58 2010, kiwamu, Update, IOO, Re:checking out & closing the vacuum chambers
Mon Dec 20 22:56:20 2010, Jenne, Update, IOO, Plan for closing the vacuum chambers
Mon Dec 20 22:56:20 2010
In reply to:
Plan for closing the vacuum chambers
If still everythig is fine, put the BS heavy door.
- Do the following list for all of the testmass chambers.
Check if the OSEMs and the OPLEV are still fine. (ITMX and ITMY were not done on Monday, so need extra care.)
Inspect the surface of the mirror with a laser pointer or a fiber coupled halogen light.
Blow the mirrors by the ionization gun.
Inspect the mirror surface again.
Move the suspension tower close to the door.
Make a single drag-wipe with iso
Move the SOS tower at the original place.
Check the OSEMs and the OPLEVs. Adjust the alignment.
Put the heavy door.
- Start slow pumping