1) True. My bad. In my elog entry (but not in my fit code) I forgot the impedance Z= 750Ohm (as in the fit) of the resonant circuit in front of the square root: Vn = Vdn + Z * sqrt( 2 e Idc )
2) That is exactly the point I was raising! The measured dark noise at resonance is 2x what I expect.
3) I don't have uncertainties for the fit offset (that is, for the Dark Noise). The quick fit that I used (Matlab's Non Linear Least Squares method) doesn't provide 95% confidence bounds when I constrain the offset parameter the way I did (I forced it to be strictly positive).
Sure. It's not a very good fit. I just wanted to see how the data was going.
I also admitted that the data points were few, especially around the 3dB point.
Today I'm going to repeat the measurement with a new setup that lets me tune the light intensity more finely.
