Hey, this looks nice, but can you provide us the comparison of rad/V with the resonant EOM of New Focus?
I have made a prototype circuit of the triple resonant EOM.
The attached is the measured optical response of the system.
The measured gains at the resonances are 8.6, 0.6 and 7.7 for 11MHz, 29.5MHz and 55MHz respectively.
I successfully got nice peaks at 11MHz and 55MHz. In addition resultant optical response is well matched with the predicted curve from the measured impedance.
However there is a difference from calculated response (see past entry) (i.e. more gains were expected)
Especially for the resonance of 29.5MHz, it was calculated to have gain of 10, however it's now 0.6. Therefore there must a big loss electrically around 29.5MHz.
I am going to re-analyze the impedance and model the performance in order to see what is going on.