I ramped the MZ PZT (with the loop disabled on the input switch) to calibrate it. Since the transmission has been blocked, I used the so-called "REFL" port of the MZ to do this.
The dark-to-dark distance for the MZ corresponds to 2 consecutive destructive interferences. Therefore, that's 2 pi in phase or 1 full wavelength of length change in the arm with the moving mirror.
Eyeballing it on the DTT plot (after lowpassing at 0.1 Hz) and using its cursors, I find that the dark-to-dark distance corresponds to 47.4 +/- 5 seconds.
So the calibration of the MZ PZT is 88 +/- 8 Volts/micron.
Inversely, that's a mean of 12 nm / V.
why am I calibrating the MZ? Maybe because Rob may want it later, but mainly because Koji won't let me lock the IFO.
Apparently, we haven't had a fast channel for any of the MZ board. So I have temporarily hooked it up to MC_DRUM at 21:13 and also turned down the HEPA. Now, let's see how stable the MZ and PMC really are overnight.
EDIT: it railed the +/- 2V ADCwe have so I put in a 1:4 attenuator via Pomona box. The calibration of MC_DRUM in terms of MZ_PZT volts is 31.8 cts/V.
So the calibration of MC_DRUM1 in meters is: 0.38 nm / count