Pointing stability of 4 days. Initial pointing does not go through suspended optics. It is launched right after the Piezo Jena steering mirrors in the BS chamber.
IP-ANG on epics screen is C1:ASC-IBQPD_X and Y in dataviewer were recentered. This beam is clipping a bit in ETMX chamber pick off mirror.
IP-POS pick off is in the BS chamber and it's qpd on the BS_ISCT This beam is also clipping just a little bit. This is easy to fix. We'll have to remove an iris from the BS optical levers table.
note: arms were not locked when I recentered
IP-ANG clipping can be traced back to our last vent of Aug. 18, 2008 See elog entry #845
This was an after earth quake - sus repair vent