It turned out that half an hour was too long. In less than that the reference cavity temperature passed the critical point when the temperature controller (located just below the ref cav power supply in the same rack) disables the input power to the reference cavity power supply.
The controller's display in the front shows two numbers. The first goes with the temperature of the reference cavity; the second is a threshold set for the first number. The power supply gets enabled only when the first number comes under the threshold value.
Now the cavity is cooling down and it will take about another hour for its temperature to be low enough and for the heater power supply to be powered.
The cavity temp cooled below SP2 set point 0.1 The Minco SP1 (present temp in Volts) now reading -0.037 so DC power supply was turned on and set to 12V 1A