Here we trended also the PMC and the MZ. The drop in the PMC happens at the same rate as the MOPA's.
That let us think that the FSS transmitteed power has gone down because of the reference cavity progressive misalignment to the laser beam.
We need to adjust that alignment sometime.
The drop in the NPRO output power (upper row, 3rd plot: Ch10 C1:PSL_126MOPA_126MON) accompained an increase of "fuzziness" in PMCTRANSPD and both coincided in time with the day we tempoarirly removed the flap from the laser chiller's chiller (July 14 2009).