I tilted the periscope beam and aligned the MC. Now the spot at the Faraday entrance is near the center of the aperture in up/down space. The arm powers are only going up to ~0.8, though. Maybe we should try a little bit of left/right.
I looked at the IP POS spot with a viewer card, and it looked round, so no obvious egregious clipping in the Faraday. Someone might take a picture with one of the GigE camera and get us a beam profile there.
We no longer have an MC1 and MC3 camera view.
I can see a bright scatterer that can be seen from the east viewport of the BSC, but I can't tell what it is. It could be a ghost beam.
It would be nice to get an image looking into the north viewport of the IOO chamber. I can't see in there because the BS oplev table is in the way.