Paco and I attempted to calculate the input matrices from May 24, 2022, but the OSEM data was all saturated and not very useful. Therefore, we decided to manually investigate the appropriate coil offsets for all BHD SUS. Before, the default offset kick was 30000 counts, but we found that LO1, AS1, AS4, and PR2 cannot take more than 5000 counts. As for LO2, SR2, and PR3 cannot take more than 2000 counts before saturating. Note that all these kick test were taken by kicking OSEM UL on all BHD Optics.
We started the freeSwing.py script on tmux freeSwing session for tomorrow at 1:00 am for only the 5000 count offset SUS.