[Yuta, Anchal, Paco]
As described briefly by JC, there were multiple failure modes going during this work segment.
Indeed, the 64 pin crimp cable from the gold sat amp box broke when work around ITMX chamber was ongoing. We found the right 64 pin head replacement around and moved on to fix the connector in-situ. After a first attempt, we suddenly lost all damping on vertex SUS (driven by these old sat amp electronics) because our c1susaux acromag chassis stopped working. After looking around the 1x5 rack electronics we noted that one of the +- 20 VDC Sorensens were at 11.6 VDC, drawing 6.7 A of current (nominally this supply draws over 5 Amps!) so we realized we had not connected the ITMX sat amp correctly, and the DC rail voltage drop busted the acromag power as well, tripping all the other watchdogs ...
We fixed this by first, unplugging the shorted cable from the rack (at which point the supply went back to 20 VDC, 4.7 A) and then carefully redoing the crimp connector. The second attempt was successful and we restored the c1susaux modbusIOC service (i.e. slow controls).
As we restored the slow controls, and damped most vertex suspensions, we noticed ITMY UL and SD osems were reading 0 counts both on the slow and fast ADCs. We suspected we had pulled some wires around when busy with the ITMX sat amp saga. We found that Side OSEM cLEMO cable was very loose on the whitening board. In fact, we have had no side osem signal on ITMY for some time. We fixed this. Nevertheless the UL channel remained silent... We then did the following tests:
DO NOT TRUST THE SATELLITE BOX TESTER 2.