Joe and Steve
The retrofitted Osaka 390 was installed on the pumpspool yesterday.
V1 gate valve is disabled for safety by disconnected pneumatic power plug.
The foreline of this maglev now have a KF25 size viton o-ring directly on the turbo.
This is bad for leak hunting.
Joe is ready with new interface cable. Power supply and cables are in place.
The maglev was pumped down this morning.
All new gas kits and metal hose were leak checked by sprayed methanol.
There is no obvious sign of leak. I was expecting the pressure to drop below 1e-5 Torr in one hour.
TP2 is drying out the levitating coils of the turbo at ~7 l/s for N2
We'll start the pump as soon as Joe is in.