I drained the water and removed side covers from the Neslab RTE 140 refrigerated water cooler unit this morning. The hoses to the laser were disconnected.
This abled you to see the little window of refregerant R404A was free of bubles, meaning: no recharge was needed.
The circulator bath was refilled with 7 liters of Arrowhead distilled water and the unit was turned on.
The water temp was kept 20.00+- .05C without any load. Finally the AC-repair man Paul showed up.
He measured the R404A level to be as specified: 23-24 PSI on the suction side and 310 PSI on the discharge side.
The unit was working fine. Paul found an intermittently functioning starting capacitor on the compressor that was removed.
The 240 micro Farad 120VAC cap will arrive tomorrow