steve, rob, alberto
Steve installed two rotary flow meters into the MOPA chiller system--one at the chiller flow output and one in the NPRO cooling line. After some hijinks, we discovered that the long, insulated chiller lines have the same labels at each end. This means that if you match up the labels at the chiller end, at the MOPA end you need switch labels: out goes to in and vice-versa. This means that, indubitably, we have at some point had the flow going backwards through the MOPA, though I'm not sure if that would make much of a difference.
Steve also installed a new needle valve in the NPRO cooling line, which works as expected as confirmed by the flow meter.
We also re-discovered that the 40m procedures manual contains an error. To turn on the chiller in the MOPA start-up process, you have to press ON, then RS-232, then ENTER. The proc man says ON, RS-232, RUN/STOP.
The laser power is at 1.5W and climbing.