I checked out what happened on c1vac. There are actually two independent monitoring codes running:
The interlocks did not trip because the low-pressure delivery line, downstream of the dual-tank regulator, never fell below the minimum pressure to operate the valves (65 PSI). This would have eventually occurred, had Jordan been slower to replace the tanks. So I see no problem with the interlocks.
On the other hand, the N2 mailer should have sent an email at 2021-04-18 15:00, which was the first time C1:Vac-N2T1_pressure dropped below the 600 PSI threshold. N2check.log shows these pressures were recorded at this time, but does not log that an email was sent. Why did this fail? Not sure, but I found two problems which I did fix:
The code then ran fine for me when I retested it. I don't see any further issues.
Installed T2 today, and leaked checked the entire line. No issues found. It could have been a bad valve on the tank itself. Monitored T2 pressure for ~2 hours to see if there was any change. All seems ok.
When I came into the lab this morning, I noticed that both N2 tanks were empty. I had swapped one on Friday (4-16-21) before I left the lab. Looking at the logs, the right tank (T2) sprung a leak shortly shortly after install. I leak checked the tank coupling after install but did not see a leak. There could a leak further down the line, possibly at the pressure transducer.
The left tank (T1) emptied normally over the weekend, and I quickly swapped the left tank for a full one, and is curently at ~2700 psi. It was my understanding that if both tanks emptied, V1 would close automatically and a mailer would be sent out to the 40m group. I did not receive an email over the weekend, and I checked the Vac status just now and V1 was still open.
I will keep an eye on the tank pressure throughout the day, and will try to leak check the T2 line this afternoon, but someone should check the vacuum interlocks and verify.