Since it seems like the entire electronics chain has no obvious failure, I decided to compensate for the apparent increased optical gain by turning the flat whitening gain down from +18dB to 0dB. Then, after some fiddling around with alignment, settings etc, I was able to lock the PRMI once again, with the ETMs misaligned, using REFL55_I to sense PRCL, and REFL55_Q to sense MICH. Some sensing matrices attached. Some notes:
So there is clearly something funky with the nominal MICH actuation scheme (MICH suspension, PRM suspension or both), which we should get to the bottom of before trying any low noise locking. I think using the ITMs as the MICH actuator in the full lock will not be a good low nosie strategy, as we would then be "polluting" all our suspended optics with our control loops, which seems highly suboptimal for technical noise sources like coil driver noise etc.