I want to lock the PRFPMI again (to commission AS WFS). Have had some success - but in doing characterization, I find that the REFL port sensing is completely messed up compared to what I had before. Specifically, MICH and PRCL DoFs have no separation in either the 1f or 3f photodiodes.
I did make considerable changes to the RF source box, and so now the relative phase between the 11 MHz and 55 MHz signals is changed compared to what it was before. But do we really expect any effect even in the 1f signal? I am not able to reproduce this effect in simulation (Finesse), though I'm using a simplified model. I attach two sensing matrices to illustrate what i mean: