Since we will have several new 1U / 2U aLIGO style electronics chassis installed in the racks, it is desirable to have a more compact power distribution solution than the fusable terminal blocks we use currently.
I did a quick walkaround of the lab and the electronics rack today. I estimate that we will need 5 units of the 24 V and 5 units of the 18 V power strips. Each end will need 1 each of 18 V and 24 V strips. The 1Y1/1Y2/1Y3 (LSC/OMC/BHD sus) area will be served by 1 each 18 V and 24 V. The 1X1/1X2 (IOO) area will be served by 1 each 18 V and 24 V. The 1X5/1X6 (SUS Shadow sensor / Coil driver) area will be served by 1 each of 18 V and 24 V. So I think we should get 7 pcs of each to have 2 spares.
Most of the chassis which will be installed in large numbers (AA, AI, whitening) supports 24V DC input. A few units, like the WFS interface head, OMC driver, OMC QPD interface, require 18V. It is not so clear what the input voltage for the Satellite box and Coil Drivers should be. For the former, an unregulated tap-off of the supply voltage is used to power the LT1021 reference and a transistor that is used to generate the LED drive current for the OSEMs. For the latter, the OPA544 high current opamp used to drive the coil current has its supply rails powered by again, an unregulated tap-off of the supply voltage. Doesn't seem like a great idea to drive any ICs with the unregulated switching supply voltage from a noise point of view, particularly given the recent experience with the HV coil driver testing and the PSRR, but I think it's a bit late in the game to do anything about this. The datasheet specs ~50 dB of PSRR on the negative rail, but we have a couple of decoupling caps close to the IC and this IC is itself in a feedback loop with the low noise AD8671 IC so maybe this won't be much of an issue.
For the purposes of this discussion, I think both Satellite Amp and Coil Driver chassis can be driven with +/- 24 V DC.
On a side note - after the upgrade will the "Satellite Amplifiers" be in the racks, and not close to the flange as they currently are? Or are we gonna have some mini racks next to the chambers? Not sure what the config is at the sites, and if the circuits are designed to drive long cables.