Now that the new Agilent full-range gauges (FRGs) have been received, I'm putting together an installation plan. Since my last planning note in Sept. (ELOG 15577), two more gauges appear to be malfunctioning: CC2 and PAN. Those are taken into account, as well. Below are the proposed changes for all the sensors in the system.
For replacements, I recommend we consider the Agilent FRG-700 Pirani Inverted Magnetron Gauge. It uses dual sensing techniques to cover a broad pressure range from 3e-9 torr to atmosphere in a single unit. Although these are more expensive, I think we would net save money by not having to purchase two separate gauges (Pirani + hot/cold cathode) for each location. It would also simplify the digital controls and interlocking to have a streamlined set of pressure readbacks.
For controllers, there are two options with either serial RS232/485 or Ethernet outputs. We probably want the Agilent XGS-600, as it can handle all the gauges in our system (up to 12) in a single controller and no new software development is needed to interface it with the slow controls.