40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop | ||||||||||

| ||||||||||

| ||||||||||

| ||||||||||

We explore bilinear SRCL to DARM noise coupling mechanisms, and show two cases that by doing BHD readout the noise performance can be improved. In the first case, the bilinear piece is due to residual DHARD motion (see also LHO:45823), and it matters mostly for the low-frequency (<100 Hz) part, and in the second piece the bilinear piece is due to residual SRCL fluctuation and it matters mostly for the a few x 100 Hz part. Details are below: ================================================= General Model: We can write the SRCL to DARM transfer function as (Evan Hall's thesis, eq. 2.29) Z_s2d(f) = C_lf(f) * F^2 * x_D + C_hf(f) * F * dphi_S * x_D ---- (1) where C_lf ~ 1/f^2 and C_hf ~ f are constants at each frequency unless there are major upgrades to the IFO, F is the finesse of the arm cavity which depends on the alignment, spot position on the TMs, etc., dphi_S is the SRCL detuning (wrt the nominal 90 deg value), x_D is the DC DARM offset. The linear part of this can be removed with feedforward subtractions and it is the bilinear piece that matters, which reads dZ_s2d = C_lf * <F>^2 * dx_D + C_hf * <F> * <dphi_S> * dx_D + 2C_lf * <F> * <x_D> * dF + C_hf * <dphi_S> * <x_D> * dF + C_hf * <F> * <x_D> * d(dphi_S). ---- (2) The first term in (2) is due to residual DARM motion dx_D. This term does not depends on the DC value of DARM offset <x_D> and thus does not depend on doing BHD or DC readout. On the other hand, the typical residual DARM motion is 1 fm << 1 pm of DARM offset. Since the current feedforward reduction factor is about 10 (see both Den Martynov's thesis and Evan Hall's thesis), clearly we are not limited by the residual DARM motion. The second term is due to the change in the arm finesse, which can be affected by, e.g., the alignment fluctuation (both increasing the loss due to scattering into 01/10 modes and affecting the spot positon and hence changing the losses), and is likely to be the reason why we see the effect being modulated by DHARD. The last term in (2) is due to the residual SRCL fluctuation and is important for the ~ a few x 100 Hz band. ================================================= DHARD effects. As argued above, the DHARD affects the SRCL -> DARM coupling as it changes the finesse in the arm cavity (through scattering into 01/10 modes; in finesse we cannot directly simulate the effects due to spot hitting a rougher location). Since in the second term of eq. (2) the LF part depends on the DARM DC offset <x_D>, this effect can be improved by going from DC readout to BHD. To simulate it in finesse, at a fixed DARM DC offset, we compute the SRCL->DARM transfer functions at different DHARD offsets, and then numerically compute the derivative \partial Z_s2d / \partial \theta_{DH}. Then multiplying this derivative with the rms value of DHARD fluctuation \theta_{DH} we then know the expected bilinear coupling piece. The result is shown in the first attached plot. Here we have assumed a flat SRCL noise of 5e-16 m/rtHz for simplicity (see PRD 93, 112004, 2016). We do not account for the loop effects which further reduces the high frequency components for now. The residual DHARD RMS is assumed to be 1 nrad. In the first plot, from top to bottom we show the SRCL noise projection at different DARM DC offsets of (0.1, 1, 10) pm. Since the DHARD alignment only affects the arm finesse starting at quadratic order, it thus matters what DC offset in DHARD we assume. In each pannel, the blue trace is for no DC offset in DHARD and the orange one for a 5 nrad DC offset. As a reference, the A+ sensitivity is shown in grey trace in each plot as a reference. We can see if there is a large DC offset in DHARD (a few nrad) and we still do DC readout with a few pm of DARM offset, then the bilinear piece of SRCL can still contaminate the sensitivity in the 10-100 Hz band (bottom panel; orange trace). On the other hand, if we do BHD, then the SRCL noise should be down by ~ x100 even compared to with the top panel. (A 5 nrad of DC offset in DHARD coupled with 1 nrad RMS would cause about 0.5% RIN in the arms. This is somewhat greater than the typically measured RIN which is more like <~ 0.2%. See the second plot). ================================================= SRCL effect. Similarly we can consider the SRCL->DARM coupling due to residual SRCL rms. The approach is very similar to what we did above for DHARD. I.e., we compute Z_s2d at fixed DARM offset and for different SRCL offsets, then we numerically evaluate \partial Z_s2d / \partial dphi_S. A residual SRCL rms of 0.1 nm is then used to generate the projection shown in the third figure. Unlike the DHARD effect, the bilinear SRCL piece does not depend on the DC SRCL detuning (for the 50-500 Hz part). It does still depends on the DARM DC offset and therefore could be improved by BHD. Since we do not include the LP of the SRCL loop in this plot, the HF noise at 1 kHz is artifical as it can be easily filtered out. However, the LP will not be very strong around 100-300 Hz for a SRCL UGF ~ 30 Hz, and thus doing BHD could still have some small improvements for this effect. | ||||||||||

| ||||||||||

| ||||||||||

| ||||||||||