Gautam showed me how the PMCTRANSPD signal was reading zero, and he suspected it might have to do with the acromag wiring. Disconnected the acromag box underneath the PSL table and checked the ADC wiring. Side note: When benchtesting the c1psl acromag chassis there was excess noise in the AI channels, and grounding the minus pin of the ADC channel eliminates the noise.
So I grounded the (-) pins on the ADC1 (192.168.113.122), which PMCTRANSPD is connected to and that seemed to fix the problem. As of right now PMCTRANSPD is reading ~.75 V.
See attached pictures
gautam: While this fix seems to have worked, I wonder why this became necessary only in the last month. Note that the problem was a noisy readback on the PMC transmission PD, which also made the FSS_RMTEMP channel noisy, leading me to suspect some kind of ground loop issue.