For a few days, I've noticed that the PSL overview StripTool panel shows PMC transmission and FSS RMTEMP channels with variation that is too large to be believable. Looking at these signals on an oscilloscope, there was no such fuzziness in the waveform. I ruled out flaky connections, and while these are the only two channels currently being acquired by the temporary Acromag setup underneath the PSL enclosure, the Acromags themselves are not to blame, because once I connected a function generator to the Acromag instead of the PMC transmission photodiode, both channels are well behaved. So the problem seems to be with the PMC transmission photodiode, perhaps a grouding issue? Someone please fix this.