While working on recovering interferometer alignment, I noticed that the ETMX Oplev SUM channel reported 0 counts. Attachment #1 shows the 200 day trend - despite the missing data, the accelerating downward decay is evident. I confirmed that there is no light coming out of the HeNe by walking down to EX. The label on the HeNe says it was installed in March 2017, so the lifetime was ~30 months. Seems a little short? I may replace this later today.