I want to collect some data with the arms locked to investigate the possibility/usefullness of having seismic feedforward implemented for the arms (it is already known to help the IMC length and PRC angular stability at low frequencies). To facilitate diagnostics I modified the file /users/Templates/Seismic/Seismic_vs_TRXTRYandMC.xml to have the correct channel names in light of Lydia's channel name changes in 2016. Looking at the coherence data, the alignment of the cartesian coordinate system of the Seismometers at the ends and the global interferometer coordinate system can be improved.
I don't know if for the MISO filter design if there is any difference in using TRX/TRY as the target, or the arm length control signal.
Data collection started at 1249018179. I've setup a script running in a tmux shell to turn off the LSC enable in 2 hours.