Rana and I talked about some (genius) options for the large range DC bias actuation on the SOS, which do not require us to supply high-voltage to the OSEMs from outside the vacuum.
What we came up with (these are pretty vague ideas at the moment):
For the thermal option, I remembered that (exactly a year ago to the day!) when we were doing cavity mode scans, once the heaters were turned on, I needed to apply significant correction to the DC bias voltage to bring the cavity alignment back to normal. The mechanism of this wasn't exactly clear to me - furthermore, we don't have a FLIRcam picture of where the heater radiation patter was centered prior to my re-centering of it on the optic earlier this year, so we don't know what exactly we were heating. Nevertheless, I decided to look at the trend data from that night's work - see Attachment #1. This is a minute trend of some ETMY channels from 0000 UTC on 18 July 2018, for 24 hours. Some remarks:
Also see this elog by Terra.
Attachment #2 shows the results from today's heating. I did 4 steps, which are obvious in the data - I=0.6A, I=0.76A, I=0.9A, and I=1.05A.
In science, one usually tries to implement some kind of interpretation. so as to translate the natural world into meaning.