After this activity, the DC bias voltage required on ETMX to restore good X arm cavity alignment has changed by ~1.3 V. Assuming a full actuation range of 30 mrad for +/- 10 V, this implies that the pitch alignment of the stack has changed by ~2 mrad? Or maybe the suspension wires shifted in the standoff grooves by a small amount? This is ~x10 larger than the typical change imparted while working on the table, e.g. during a vent.
Main point is that this kind of range requirement should probably be factored in when thinking about the high-voltage coil driver actuation.
We unstuck ETMX by shaking the stack. Most effective was to apply large periodic human sized force to the north STACIS mounts.