Now that the IMC is remaining locked for extended periods of time, the next problem to attack is the ASS dither alignment system. For a start, I decided to try and get the POX and POY locking working again, as we have not fully recovered the interferometer alignment after the most recent pumpdown. I spent a couple of hours tweaking the alignment of the arm cavity mirrors, BS, and TTs to try and recover the maximum possible TRX and TRY - however, my best efforts only yielded TRX~0.8, TRY~0.75. Moreover, the beam axis is such that the spot is significantly off in YAW on both ETMs, as evidenced by the camera views (also true but less obvious on the ITMs). However, trying to bring the beam back to the center of the optics yields TRY and TRX values lower than the above reported maxima. The EX green beam is currently unavailable to verify the arm cavity alignment because of my hijacking the EX NPROs PZT control for PLL investigations, but with the Y arm, I'm able to lock a TEM00 mode. Probably just needs more careful systematic alignment, but I'm not pursuing this tonight.