At ~930am, I vented the IY annulus by opening VAEV. I checked the particle count, seemed within the guidelines to allow door opening so I went ahead and loosened the bolts on the ITMY chamber.
Chub and I took the heavy door off with the vertex crane at ~1015am, and put the light door on.
Diagnosis plan is mainly inspection for now: take pictures of all OSEM/magnet positionings. Once we analyze those, we can decide which OSEMs we want to adjust in the holders (if any). I shut down the ITMY and SRM watchdogs in anticipation of in-chamber work.
Not related to this work: Since the annuli aren't being pumped on, the pressure has been slowly rising over the week. The unopened annuli are still at <1 torr, and the PAN region is at ~2 mtorr.