When all things fail (netgpibdata.py is giving me weird data. When I plot the data it has saved from the 4395A, it's some wierd other universe's version of my transfer function. I don't really know what's up. I'm pretty sure I'm getting the 'correct' data, since each TF looks vaguely like it should, but with some crazy humps. I'll talk to Yoichi in the morning about it maybe.) (also, we're low on emergeny floppy discs), you can always take a picture of the Agilent 4395's screen, as shown below.
* Mode cleaner and PMC are both relocked after my shenanigans, and I'll try again in the morning (I assume locking is going on tonight) to get real TF's with real data, as opposed to the photo method.
Note to self: post the data of the TFs in the elog along with the plots, for posterity.
These TFs are of the Mode Cleaner servo board, exciting IN1 (or the 3.7MHz notch pomona box which is connected to IN1), and measuring at the SERVO out of the board.
One with the box, one without the box, and one of just the box for good measure.