Not logged in
Wed Nov 28 21:51:52 2007, tobin, Configuration, PSL, ISS
Thu Nov 29 15:17:53 2007, rob, Configuration, PSL, ISS
Mon Dec 3 22:20:09 2007, tobin, Configuration, PSL, ISS
Thu Nov 29 15:17:53 2007
In reply to:
Reply to this:
I put some ISS beam on the diode on the PSL table. In the previous layout, this was the monitor diode (and it's labeled monitor) but I plugged it into the sensor jack anyways so we can run with the loop closed for now; we can just switch the cables later. The reason the beam was unclear is because someone popped up a flipper mirror which redirects the beam from the ISS into an OSA.
With the ISS gain slider at 15 dB the UGF is around 40kHz.
Why do we have such short cables for the ISS diodes?