As discussed at the meeting earlier this week, we will use some old *MOPA* channels for interfacing with the PLL system Jon is setting up. He is going to put a sketch+photos up here shortly, but in the meantime, Koji helped me identify a channel that can be used to tune the temperature of the Lightwave NPRO crystal via front panel BNC input. It is C1:PSL-126MOPA_126CURADJ, and is configured to output between +/-10V, which is exactly what the controller can accept. The conversion factor from EPICS value to volts is currently set to 1 (i.e. EPICS value of +1 corresponds to +1V output from the DAC). With the help of the wiring diagram, we identified pins 3 and 4 on cross-connect #J7 as the differential outputs corresponding to this channel. Not sure if we need to also setup a TTL channel for servo ENABLE/DISABLE, but if so, the wiring diagram should help us identify this as well.
The cable from the DAC to the cross-connect was wrongly labelled. I fixed this now.