To test the hypothesis that the IMC lock duty cycle is affected by the PRM alignment. Rana pointed out today that the input faraday has not been tuned to maximize the output->input isolation in a while, so the idea is that perhaps when the PRM is aligned, some of the reflected light comes back towards the PSL through the Faraday and hence, messes with the IMC lock.
I've made a simple script - the pseudocode is the following:
The idea is to keep looping the above over the weekend, so we can expect ~100 datapoints, 50 each for PRM misaligned/aligned. The times at which PRM was aligned/misaligned is also being logged, so we can make some spectrograms of PC drive RMS (for example) with PRM aligned/misaligned. The script lives at /opt/rtcds/caltech/c1/scripts/SUS/FaradayIsolationTest/FaradayIsolCheck.py. Script is being run inside a tmux session on pianosa, hopefully the machine doesn't crash over the weekend and MC1/CDS stays happy.
A more direct measurement of the input Faraday isolation can be made by putting a photodiode in place of the beam dump shown in Attachment #1 (borrowed from this elog). I measured ~100uW of power leaking through this mirror with the PRM misaligned (but IMC locked). I'm not sure what kind of SNR we can expect for a DC measurement, but if we have a chopper handy, we could put a chopper (in the leaked beam just before the PD so as to allow the IMC to be locked) and demodulate at that frequency for a cleaner measurement? This way, we could also measure the contribution from prompt reflections (up to the input side of the Faraday) by simply blocking the beam going into the vacuum. The window itself is wedged so that shouldn't be a big contributor.