Yesterday, while we were bringing the CDS system back online, we noticed that the control room wall StripTool traces for the seismic BLRMS signals did not come back to the levels we are used to seeing even after restarting the PEM model. There are no red lights on the CDS overview screen indicative of DAQ problems. Trending the DQ-ed seismometer signals (these are the calibrated (?) seismometer signals, not the BLRMS) over the last 30 days, it looks like
I poked around at the electronics rack (1X5/1X6) which houses the 1U interface box for these signals - on its front panel, there is a switch that has selectable positions "UVW" and "XYZ". It is currently set to the latter. I am assuming the former refers to velocities in the xyz directions, and the latter is displacement in these directions. Is this the nominal state? I didn't spend too much time debugging the signal further for now.