Between frequent MC1 excursions, I worked on ALS recovery today. Attachment #1 shows the out-of-loop ALS noise as of today evening (taken with arms locked to IR) - I have yet to check loop shapes of the ALS servos, looks like there is some tuning to be done.
On the PSL table:
Note: Some of the ALS scripts are suffering from the recent inablilty of cdsutils to pull up testpoints (e.g. the script that is used to set the UGFs of the phase tracker servo). The workaround is to use DTT to open the test points first (just grab 0.1s time series for all channels of interest). Then the cdsutils scripts can read the required channels (but you have to keep the DTT open).