I installed a DC PD (Thorlabs PDA 520) in the beam path to AS55. I placed a 2" 90/10 BS on a flip mount that picks of enough light for the PD to spit out ~8V when the port is bright. Single arm continuous signal will be ~2V. While most of the light still continues towards AS55, the displacement from the BS moves the beam off AS55, so I used the flip mount in case anyone needs to use AS55. The current configuration is UP.
When we're done with loss investigations the flip mount should be removed from the bench.
I hooked the PD up to an ethernet-enabled scope and started scripting the loss map measurement (scope can receive commands via http so we can automate the data acquisition). The scope that was present at the bench and had been used for the MC ringdown measurements had a 'scrambled' screen that I couldn't fix so I had to retrieve another scope ("scope1"). I'll try to find out what's wrong with it but we may have to send it in for repair.