We're waiting on the last couple electrical components to arrive that are needed to complete the acromag chassis, but it is essentially operational. Right now it is connected to the PSL Mephisto's diagnostics port, for which only a single XT1221 A/D unit is needed. We assigned the IP address 192.168.113.121 to it. For the time being I'm running a tmux session on megatron (named "acromag") that grabs and broadcasts the epics channels, with Lydia's original channel definitions. Since the chassis is 4U tall, there's not really any place in the rack for it, so we might want to move it to the X-end before we start shuffling rack components around. Once we finalize its location we can proceed with adding the channels to the frames.
For the eventual gradual replacement of the slow machines, we need to put some thought into the connectors we want in the chassis. If we want to replicate the VME crate connectors we probably need to make our own PCB boards for them, as there don't seem to be panel-mount screw terminal blocks readily available for DIN 41612 connectors. Furthermore, if we want to add whitening/AA filters, the chassis may actually be large enough to accomodate them, and arranging things on the inside is quite flexible. There are a few things to be considered when moving forward, for example how many XT units we can practically fit in the chassis (space availability, heat generation, and power requirements) and thus how many channels/connectors we can support with each.
Steve: 1X3 has plenty of room