P1 IFO pressure is 1 mTorr, valve configuration: vacuum normal, annulosses are pumped, RGA is off, not pumped.
THANKS to Chris !
The shutters can be opened with high power.
No communication error message still exist.
I will reboot c1vac1 and c1vac2 to get gauge communication with medm tomorrow.
The pumpdown had stalled because of some ancient vacuum interlock code that prevented opening the valve V1 between the turbo pump and the main volume.
This interlock  compares the channels C1:Vac-P1_pressure and C1:Vac-PTP1_pressure, neither of which is functioning at the moment. The P1 channel apparently stopped reading sometime during the vent, and contained a value of ~700 torr, while the PTP1 channel contained 0. So the interlock code saw this huge apparent pressure difference and refused to move the valve.
To bypass this check, we used caput to enter a pressure of 0 for P1.