Not logged in
Thu Jan 22 12:08:59 2009
Following the procedure described in Royal Reinecke's 2006 SURF report, I've calibrated the ETMY yaw oplev DOF. The idea is to sweep the mirror tilt, measuring the transmitted cavity power and the oplev error signal. The cavity power can be related to the mirror tilt in radians following D. Anderson APPLIED OPTICS, Vol. 23, No. 17, 1984.
I've made a simple matlab script which spits out the final number; it calls Royal's perl script to do the sweep. I get 420 microrad/ct for ETMY yaw. In 2006 Royal got 250 microrad/ct. Could something have changed this much, or is one of us wrong? I'll double check my procedure and do the other arm cavity oplevs, and describe it in detail when I have more confidence in it.
Kakeru and I plan to extend this to handle the PRM, SRM, and BS. One script to rule them all.