Not logged in
Fri Jan 16 18:33:54 2009
c1lsc rebooted to fix 16Hz glitches
There were 16Hz harmonics in the PD3 and PD4 channels even when there is no light falling on it.
Actually, even when the connection to the ADC was removed, the 16Hz noise was still there.
Rob suggested that this might be digital problem, because data is sent to the daq computer very 1/16 of a second.
We restarted c1lsc and the problem went away.