Using the ALS green beat and armlength feedback I mapped an IR resonance of the Y-Arm by stepping through a ramp of offset values.
First I optimized the IR alignment with the dither scripts while LSC kept the arm on resonance, and then transitioned the length control to ALS. The beat frequency I obtained between the Y-arm green and the PSL was about 25 MHz. Then I applied a controlled ramp signal (stepping through small offset increments applied to LSC-ALSY_OFFSET, while logging the readback from channels LSC-TRY_OUT16 and ALS-Y_FC_SERVO_INMON with an averaging time of 1s.
The plots show the acquired data with fits to and , respectively.
The fits, weighted with inverse rms uncertainty of the data points as reported by the cds system, returned HWHM = 0.6663 ± 0.0013 [offset units] and m = -0.007666 ± 0.000023 [MHz/offset unit], which gives a combined FWHM = 10,215 ± 36 Hz. The error is based purely on the fit and does not reflect uncertainties in the calibration of the phase tracker.
This yields a finesse of 388.4 ± 1.4, corresponding to a total loss (including transmissivities) of 16178 ± 58 ppm. These uncertainties include the reported accuracies of FSR and phase tracker calibration from elog 9804 and elog 11761.
The resulting loss is a little lower than that of elog 11712, which was done before the phase tracker re-calibration. Need to check for consistency.