Not logged in
Mon Dec 8 17:50:21 2008
The suspension drift monitor shows that the ITMY alignment was shifted after the earthquake.
Looks like only the UL sensor had a step at the earthquake (see the attachment 1).
So it is probably an electronics problem.
I pushed in the cable between the rack and the ITMY satellite amplifier, but no change observed.
Actually, the ITMY-UL sensor looks like it has been dead before the earthquake.
The second attachment shows a long-term trend of the UL sensor.
The sensor output had been around zero since Nov. 17th.
When I disabled the output of the UL sensor, the sus-drift-mon fields turned green.
So I think the drift-mon's reference values are wrong, and currently the ITMY is in a good alignment.
I also attached the free-swing measurements of the ITMY taken on Aug. 18th and today.
There is no notable change in the resonant frequencies.