Our online subtraction filters for PRC angle and MC length were trained on the raw ADC signals, so I've inverted the filters that Rana installed in the PEM filter banks in the OAF signal conditioning filter banks (C1:OAF-WIT_STS1X, etc.)
It's not perfect, since the inversion would be unstable, and thus needs a low pass. I used an ELP at 800Hz. The error in the inversion is then something like half a degree at 5Hz, which is the highest frequency we really ever subtract at. It should be ok.