The interferometer is warming up!
I had some issues locking the IMC at first. It turned out that the MC3 side OSEM signal wasn't getting to the ADC. A satellite box sqush fixed it.
I touched up the PMC alignment; the best I could do is 0.75V, probably due to the AOM being in place.
I haven't touched the WFS offsets, but the current ones seem to be doing ok. I'll touch them up tonight when the seismic activity has calmed.
I made some changes to the state of the PZT/PC crossover gain in the mcdown script, resulting in the IMC catching lock quicker.
Thankfully, the tip tilt pointing stayed good during the upgrade. I barely had to touch the ETM alignment to lock the arms. ETMX is showing some errant motion, though...