As discussed at today's meeting, we would like to (re)measure the Arm cavity lengths to ~mm precision, and their g-factors. Any arm length mismatch affects the reflection phase of the sidebands in the PRMI, which might be one source of our woes. Also, as I mentioned in a previous elog, the g-factors influence whether our 2f sidebands are getting pulled into the interferometer or not.
These both can be done by scanning the arm on ALS and measuring the green beat frequency at each IR resonance. (Misaligning the input beam will enhance the TM10 Mode content, and let us measure its guoy phase shift)
I started working on this today, but I have measurements to do, since at the time of today's measurements, I was fooled by the limits of the ALS offset sliders that I could only scan through two FSRs. Looking back at Manasa's previous measurment (ELOG 9804), I see now that more FSRs are possible.
Ways I will try to improve the measurement:
Just for kicks, here are scans from today.