I assembled the telescope to couple PSL light into the fiber. The maximum coupling that I could obtain was 10mW out of 65mW (~15%).
I was expecting to achieve 80-90% coupling from my design estimates. It makes me wonder if the beam waist measurements made by Harry during summer were correct in the first place. I would like to go back and check the beam waist at the PSL table.
Also, we need a pair of 8m (~25 feet) long SMA cables to carry the RF signal from the beat PD on the PSL table to frequency counter module on the IOO rack.
Steve says that we had a spool of SMA cable and it was borrowed by someone a few months ago. Any updates on either who is holding it or if it has been used up already would help.
The X end slow computer was down this morning. So I used only the Y arm ALS to record the noise level for reference. DTT data for ALSY out of loop noise before opening PSL enclosure is saved in /users/manasa/data/141211/ALSYoutLoop.xml